Quality level gauge manufacturer? Mechanical interference, the so-called mechanical interference, means that some components inside the radar level sensor will also vibrate due to the large-scale vibration or impact of the external machine, and even undergo displacement and deformation, which may also cause the pointer of the instrument head to loosen and cause measurement. error. In this case, we usually use spacers, shock springs, etc. to cushion the shock. See extra info at https://www.kaidi86.com/product.html.
For more accurate measurement in deaerators, magnetrol guided wave radar (GWR) is a preferred option. Since its performance and accuracy are not contingent on the specific gravity and/or inference, it can provide reliable measurements in all situations, including the difficult and turbulent process conditions of deaerators and feedwater heaters. In addition, GWR does not require external inputs or calibration to achieve specified performance. This effectively eliminates the introduction of errors during the calibration process or from external sources, i.e., pressure and temperature. With this high level of accuracy, operators can trust that their deaerators will be well controlled.
So what can be done about these difficulties? Under the condition of strong dust, on the one hand, the radar with high transmitting energy can be selected, on the other hand, the measurement software with continuous measurement algorithm of wave-loss waiting can be selected. When the radar encounters strong dust, it will not misjudge the measurement result even if the radar loses wave for a short time. After entering the state of continuous measurement algorithm, if the reflection wave of real material surface can be recognized within the set waiting time, the correct measurement value of material surface can be obtained. In the past, only a few foreign radars have this function. At present, there are also domestic radars with this function, and the practical application effect is very good.
Measurement accuracy, the accuracy of ordinary radar is generally ±10mm, and the accuracy of precision radar is ±3mm. Selection according to the actual needs of production. Range, according to actual needs, choose the antenna size. Note that the actual range is reduced in complex environments. Antenna type and antenna size, the larger the antenna size, the larger the measured range and the stronger the anti-interference ability. The antenna types of radar level gauge are rod type, bell mouth type, paraboloid and so on.
Measuring principle of radar water level meter: The radar level gauge adopts the working mode of transmitting-reflecting-receiving. The electromagnetic wave emitted by the antenna is reflected by the surface of the measured object and then received by the antenna. After measuring the distance from the water surface to the radar antenna, the elevation of the water surface can be calculated according to the elevation of the radar antenna. The radar water level meter adopts pulse wave technology with low power consumption. It can be powered by two-wire 24 VDC, 485 interface output, or can be directly powered by 12 VDC, SDI-12 interface output, with high accuracy and wider application range.
As one of the most professional magnetic level indicators manufacturers in China,Guangdong Kaidi Energy Technology Co., Ltd. provides customized solutions for a range of industrial automation process applications,such as mechanical float level indicator.We specialized in radar level gauge, fork type level switch,etc. These were implemented successfully, and KAIDI magnetic level gauge manufacturers products,such as radar level meter, magnetic level indicators, can be used in many different industries such as food & beverage, water, energy, pharmaceutical etc. Read even more information on kaidi86.com. Our Radar Level Meter has a range of up to 150 meters, frequency up to 120GHz and an accuracy of ±1mm, which can cope with various complex measurement conditions.
There is AC interference and the voltage is high. For example, for the radar level meter used in the production line, the power supply requirement is 24VDC (typical value), but in the on-site measurement, it is found that the power supply is displayed as 27.2V, which is significantly higher than 24VDC, resulting in a large measurement result and even a radar level meter. crash phenomenon. The installation position of the radar level meter is incorrect, which leads to deviations in the measurement. For example, the accumulation of aggregates in the transfer bin is a “mountain”-shaped cone, but only one radar level meter is installed near the discharge port of the return belt. , the installation position is too close to the discharge opening of the return belt, and at the same time, it is too far from the discharge opening of the feeding belt on both sides. Just below the radar level meter is the drop point of the return belt. If the distance is too close, the aggregate in the falling process will interfere with the radar level meter and form false reflections.
When the distance between the liquid level of the measured medium and the electromagnetic wave transmitter is less than 4m, the guided wave radar is selected. If the distance between the liquid level of the measured medium and the electromagnetic wave transmitter is greater than 4m and less than 35m, select the guided wave radar with the guided wave cable. When the distance between the liquid level of the measuring medium and the electromagnetic wave transmitter is greater than 35m, or when the liquid level of the measuring medium is high temperature, high viscosity or solid liquid level, the air-shooting radar should be selected.